What is the Breath Alcohol Tester with Fuel Cell sensor
Fuel cell breathalyzer is the 'gold standard' of hand-held alcohol tester for both personal and professional use. They are the choice of professional users such as Law Enforcement and serious personal users such as concerned parents and individuals who require measurement they can trust.
In fuel cell instruments, the breath is directed into a fuel cell with dual platinum electrodes where alcohol is oxidized and generates electrical current. The higher the alcohol content of the breath, the greater the output current of the fuel cell. This current is a direct indication of the amount of alcohol consumed by the fuel cell. By precisely measuring the current produced in the fuel cell, an accurate measure of Breath alcohol content (BAC) is recorded.
CA05FL uses the Platinum Fuel Cell Technology that has the following performance characteristics not found in semiconductor devices.
* High accuracy.
CA05FL is consistently accurate across a wide alcohol concentration range from 0.00 to 0.40 BAC. Unlike semiconductor devices, the CA05FL can report 3-digit BAC measurement accuracy because of the precision of its Platinum Fuel Cell Technology.
* Alcohol specific.
Semiconductor devices can give positive alcohol readings even when no alcohol is present. They will react to perfume, hairspray, gasoline, cigarette smoke, and breath substances such as acetone and ketones. CA05FL measurements are always and only.
* Calibration stability.
Unlike semiconductors which require frequent re-calibration, CA05FL recommends recalibration only once every six months under normal use.
* Long working life.
Unlike Semiconductor devices that have a short life span, CA05FL Fuel Cell instruments have an expected working life of 3~5 years. CA05FL fuel cell tester will be in the field and in continuous use by medical clinics and personal or other industrial measurement requirements.
The combination of long working life, calibration stability, alcohol specificity, and high precision all add up to a device that when properly maintained can be trusted to perform.
* fuel cell sensor, fuel cell alcohol detector, fuel cell alcohol tester, breathalyzer, alcohol breath tester, fuel cell alcohol breath tester, CA05fl, ca-05fs, alcohol detector,
alcohol tester at CA05fl, breathalyzer alcohol tester, digital alcohol tester,
digital alcohol meter, digital alcohol breathalyzer, ca-05fl, CA05fl alcohol
tester, alcohol breath tester, breath alcohol tester, chemical alcohol tester, chemical sensor, chemical alcohol sensor, fuel-cell breathalyzer, fuel cell sensor, fuel-cell alcohol tester,
Cell Sensor (Own developed high selective sensor)
* Fast warm-up, respond and
* Law enforcement grade accuracy (3rd decimal place display)
Easy Calibration & low cost maintenance
* Intelligent functions
Accurate sampling system
- Self-diagnosis system(Flow, Calibration,
Temperature, Low battery)
- Auto power saving
1. Insert a mouthpiece into the mouthpiece inlet
2. Push the ON/OFF power switch under the display window.
- The unit will show "Full display" along with a welcome sound. And then it starts countdown which will be taken
about 15 ~ 35 seconds. This warm-up process prepares the sensor and circuit for testing.
3. When you hear a beep sound & "bLo" display is on, please blow steadily into the mouthpiece
until you would hear the "Beep" sound again. Stop blowing.
- In this stage, if you don’t blow within 20 seconds, it automatically turns off.
Try to power on again.
4. After the circling line is on the display for 3 ~ 22 seconds, the test result(mg/l BrAC or ‰ Promile) will be displayed for 20 seconds along with beep sound.
- If the test result is over 0.25mg/l / 0.50‰ (default setting) respectively, the LCD color will changed to red along with "Alarm" sound.
- The result will be displayed for 20 seconds .(light on for the first 15 seconds and light off for another 5 seconds)
If you push the power switch during the last 5 seconds, the unit goes back to countdown stage for another test.
5. Finally, the unit will show "OFF" and turn off automatically along with a sound.
6. For the successive testing, try again from No.2 to No.4.
7. Space tests at least 2 minutes apart.
※ After using, you can turn the device off by pressing the power switch for 2 sec.